Check if the device is in advertising mode (Bluetooth symbol blinking), if your master has a compatible Bluetooth module (revision 4.0 or later) and operating system (at least Windows 7, OS X 10.9, iOS 6.0, Android 4.4).
Recent Sylvac instruments use different Bluetooth profiles (Simple, Pair, HID). An instrument in HID profile cannot be connected by a program or application. Check or change the active configuration in the BT.CFG menu. Select the Single or Pair profile for connection with Sylcom or Sylvac Anywhere.
Some older systems (e.g. Windows 7) sometimes need to update the drivers of their Bluetooth modules to ensure compatibility with Bluetooth Low Energy HID devices. Look for an updated driver package for your wireless module.
Check the keyboard layout used by your master device. Sylvac instruments are designed to work with English (US) keyboards. Other keyboard layouts might work too, but some keyboard layouts (e.g. French) do not produce correct values, e.g.+008.521 <
The values from my measuring device with HID profile are transmitted correctly but followed by additional non-numerical characters
Make sure to use an English (US) keyboard layout for best compatibility. In tolerance mode the device transmits the tolerance judgement using additional characters.
I installed the English (US) keyboard layout on my master device, but I still get strange characters from the HID device
Check the enabled keyboard layout. Some systems (e.g. Windows) can handle a different keyboard layout for each program or application; others (e.g. Android) use a keyboard layout per input device.
The values from Sylvac Bluetooth devices in HID mode are transmitted like keystrokes from an external keyboard. They are received and interpreted by the currently active application. If keystrokes are received by an application that is not designed to display text, behavior is unpredictable. Make sure to put the text input cursor in the application that should receive the value before sending it.
Several factors can influence the achievable range of a Bluetooth connection, e.g. large metallic parts, walls or other wireless devices. If you use a Bluetooth low energy dongle, do not place it next to any USB 3.0 device, this might cause interferences and considerably shorten the range of your connection.
See below for error codes, possible reasons and trouble-shooting
Err.bt 0: Current consuming connection parameters. Battery life might be reduced. Press any key to acknowledge this error.
Err.bt 1: General Bluetooth error. Disconnect and reconnect the device. Remove the pairing established in former connection from the master device(s) (smartphone, tablet or PC) and the measuring device (menu BT – Reset) if this error is displayed when trying to connect.
Err.bt 2: Bluetooth initialization error. Restart the device.
Err.bt 4: Bluetooth memory error. Disconnect and reconnect the device.
No.data: Value could not be sent via Bluetooth® Technology. Check the Bluetooth® connection on instrument and master device (on/off). Check if you are within the max. transmission range 5-15m.
No.bt: Bluetooth module is not connected. Connect to a compatible master device.
-----: Initialization in progress. Message disappears automatically. Press a button if no value is displayed after five seconds.
Err.0: Sensor error. Press SET key.
Make sur your measuring device uses Simple or Paired profile (not HID). Check for updates on www.sylvac.ch. Vmux v1.43 and Sylcom v1.3.1 or later are compatible with all Sylvac Bluetooth devices.
- Register and log in Sylvac website
- Go to menu Registration/Developer resources
- Get your download link/access to developer resources within 24h (Communication DLL, Bluetooth® profiles specifications, etc.)
- Stay updated on new versions by email notifications